- Woodpecker Endo Gold Endomotor is mainly used in Endodontic treatment.During root canal preparation procedure,it is used to mold & clean the root canal.ADVANTAGES:OLED display
a) Customized program sequence number 1-9, totally 9 programs
b) Operation mode
c) Battery consumption
d) Set speed
e) Set torque
f) File system
g) Operation mode
h) Torque setting
i) Real-time torque
Protective function of automatic reverse
During operation, if the load value exceeds the preset torque value, the file rotation mode will automatically change to Reverse Mode. And the file would return to normal rotation mode when the load is below the preset torque value again.
a) Clockwise rotation: Load value is lower than preset torque value
b) Counterclockwise rotation: Load value is higher than preset torque value.
c) Clockwise rotation: Load value is lower than preset torque value again.
- The device is suitable for shaping the root canals during root canal treatment & an advantage in replacing the hand files to rotary endodontics.
- Patients with implanted pacemakers (or other electrical equipment).
- Wireless handpiece enables more convenient operation.
- Adopt real-time feedback technology and dynamic torque control,effectively preventing needle breakage.
- Storage of 9 user-defined modes allows invocation at any time. Under each mode, Continuous Rotation Mode, Reciprocating Motion Mode, and Reverse Rotation are for options.
- TECHNICAL SPECIFICATIONS:
Battery Lithium battery in handpiece: 3.7V /1400mAhPower adapter Input: ~100V-240V 50Hz/60Hz 0.4A Max Output: DC5V/1A Torque 0.6Ncm-5.0Ncm(6mNm ~ 50mNm) Rotate speed 100rpm~1000rpm Environment temperature +5°C ~ +40°C Relative humidity 30% ~ 75% Atmospheric pressure 70kPa ~ 106kPa
- This equipment should be stored in a room where the relative humidity is 10% ~ 93%, atmospheric pressure is 70kPa to106kPa, and the temperature is -20°C ~ +55°C.
- Avoid the storage in a too hot condition. High temperature will shorten the life of electronic components, damage battery, reshape or melt some plastic.
- Avoid the storage in a too cold condition. Otherwise, when the
temperature of the equipment increases to a normal level, there will be dew that will possibly damage PCB board.
- Oiling of contra-angle:Only the original oil injection nozzle can be used for oiling of contra-angle. After disinfection of contra-angle and before sterilization, oiling should be conducted under high pressure and high temperature screw the injecting nozzle into jet of oil bottle. (Around 10 circles)
- Plug the nozzle into the end part of contra-angle, and then
grease the contra-angle for 2-3s till the oil flow out of contra-angle head
- Vertically place the end part of contra-angle or tilt the contra-angle to
let go the redundant oil under gravity.
- 1 x Endomotor Handpiece
- 1 x Charging Base
- 1 x Contra-angle
- 1 x Power Adapter
- 1 x Lubricator
- Instructional Manual
1 x Charging Base, 1 x Contra-angle, 1 x Endomotor Handpiece, 1 x Lubricator, 1 x Power adapter, Instructional Manual
There are no enquiries yet.